|Sheffield Measurement Head|
There are many reasons why the manufacturers of photocopy machines have target ranges for Sheffield roughness. A xerographic machine needs optimum paper surface properties for reliable sheet feeding, image transfer, and image fix. The fix level decreases as the Sheffield roughness increases, as it affects toner adhesion. Print density loss is observed as roughness increases. There also can be image problems with papers that are too smooth. Toner particles can be flattened and appear as larger dots, thus increasing the perception of the background. Rougher papers produce less background. With regards to paper handling, smoother papers are less stiff for a given basis weight. Smoother papers increase “electrostatic tacking” in the image transfer process. The coefficient of friction decreases with increasing roughness, a factor that is important in sheet transfer operations. The Sheffield roughness properties are carefully specified for the electrostatic copier printing process.
|PROFILE/Plus Roughness Tester|
There are some applications where metallized films are applied to the surface of paper. The reflectance properties of the film can expose wire marks on the base sheet. This is another example where the gentle loading force of Sheffield test better replicates end use properties of the paper, as compared to the PPS test.
Many plastic films are packaged in reams, like paper, for use in a photocopier to produce overhead projector transparencies. When the surfaces of the films are extremely smooth, there are static forces and cohesive forces that interfere with single sheet feeding. The manufacturers of such films generally create rough surfaces that enable an air film to exist between sheets. It is common to use Sheffield test results to control the process that generates the rough surface. Again, the PPS test would have measuring head loading that is excessive for this test.
When selecting a test instrument for paper, it is important to understand the relationship between the end-use of the product and the physical test parameters of the instrument. A further requirement is to use a test where process control settings on a paper machine (or plastic web processing equipment) can be modified to optimize the final product for its intended end use. The old adage was “If you can’t control it, why measure it?” In today’s marketplace, the customer will be able to find a supplier who makes the product he wants.
Learn more about Sheffield and PPS Roughness at the Technidyne Website.